How do I test my OneTouch® UltraMini® Meter with OneTouch® Ultra® Control Solution?
How to perform a control solution test
Start with the meter off. If you have turned the meter on to change settings or review past results, turn it off.
Check the code on the test strip vial before inserting the strip
Insert a test strip to turn on the meter
Make sure the three contact bars are facing you when you insert a test strip. Push the strip in as far as it will go. Do not bend the strip.
Match the code on the meter with the code on the test strip vial
If the code on the meter does not match the code on the test strip vial, press ▲ or ▼ to match the code number on the test strip vial. The new code number will flash on the display for three seconds, and then stay constant for three seconds. The display will advance to the screen with the flashing blood drop icon.
If the codes already match, wait three seconds. The display will advance to the screen with the flashing blood drop icon.
Mark the test as a control solution test
- Mark all control solution tests with CtL. This will stop them from being stored as blood glucose results.
- Control solution results marked with CtL are not stored in the meter’s memory.
Press ▲ so that the control solution test symbol CtL appears in the upper right corner of the display. You must mark the test before you apply control solution. Once you have completed the test, you cannot change the marking. The meter is now ready to perform a control solution test.
Prepare and apply control solution
Shake the control solution vial before each test. Remove the cap and squeeze the vial to discard the first drop. Then wipe the tip with a clean tissue or cloth. Hold the vial upside down and gently squeeze a hanging drop.
Touch and hold the hanging drop of control solution to the narrow channel in the top edge of the test strip. Make sure the confirmation window fills completely. Control solution should not be applied to the flat face of the test strip.